The Specification Of AtmoExplorer Temperature Humidity Chamber
Temperature range: -20°C~+150°C
Humidity range: 20-98%RH
Temperature and humidity control accuracy: ±0.5°C; ±3%RH
Temperature and humidity distribution uniformity: ±2°C; ±3%RH
Heating time: temperature to 100°C within 30 minutes
Cooling time: temperature down to -20°C within 60 minutes
Testable area: 500*750*600mm (D*W*H)
Various Sizes Can Be Customized: such as 80L, 150L, 225L, 408L, 800L, 1000L
ChiuVention Environmental Chamber Temperature And Humidity Control Accuracy
Temperature and humidity control precision: ±0.5°C; ±3%RH
Temperature and humidity uniformity: ±2°C; ±3.0%RH
Rapid Heating And Cooling Of The Environmental Test Chamber
When unloaded, the AtmoExplorer can achieve heating from room temperature to 100°C within 30 minutes. It can also achieve cooling from room temperature to -20°C within 60 minutes.
ChiuVention Climate Chamber Has A Wide Range Of Applications
The test chambers can simulate the environment of high temperature, low temperature, high humidity, and low humidity.
And this temperature humidity chamber is suitable for testing requirements in various fields and industries such as electronics, materials research, medicine, automobiles, food, etc.
The Constant Temperature Humidity Chambers Operates Stably:
The heating element is made of spring-loaded Nichrome wire. Solenoid valve and other refrigeration auxiliary parts are used first-line brands. France imported “Taikang” full closed type high-efficiency compressor. Humidifier for stainless steel electric steam generator in the form of UL-type humidifier.
Functional Structure Of The Climate Cabinet:
7 inch large touch screen, easy to operate.
Built-in LED window lighting, easy to observe the sample.
Equipped with RS-232, USB interface, external computer copy data.
With reservation startup and shutdown function, key and screen lock function.
With power failure program memory, automatically start and continue to execute the program function after power restoration.
380V 50HZ 20A 3φ 4 wire
225L 1330*940*1670mm (D*W*H)
Safety Protection Device Of Constant Temperature And Humidity Chamber
Anti-burn switch, no fuse switch protection.
Short-term water shortage alarm and long-term water shortage shutdown protection.
Compressor high pressure protection, compressor overheating protection, compressor overcurrent protection.
Why conduct environmental simulation tests?
Predicting Product Durability in Transit:
Environmental simulation experiments enable the simulation of diverse transit conditions (high temperature, low temperature, humidity, pressure, vibration) to accurately predict product durability during transportation.
Optimizing Product Packaging Design:
Through environmental simulation experiments, different packaging designs can be thoroughly tested to identify the most effective one, preventing damage, contamination, and other issues in the transportation process.
Enhancing Product Quality and Safety:
Environmental simulation experiments assess the performance of product packaging across various environments, mitigating risks of damage, pollution, and other issues during transportation, ultimately improving overall product quality and safety.
By conducting environmental simulation experiments, potential issues with product packaging can be identified before transportation, preventing damage, pollution, and other problems. This proactive approach helps in avoiding unnecessary losses and costs.
What benefits can environmental simulation tests bring to enterprises?
Reliability Assessment of Products:
Environmental simulation tests replicate real-world conditions, enabling a thorough assessment of product performance and reliability across diverse environmental scenarios.
Product Compliance Assessment:
These tests assist companies in evaluating if their products adhere to specific environmental standards and regulatory prerequisites, ensuring both compliance and market acceptance.
Quality Control of Products:
These tests enable companies to implement robust quality control measures during the production process, ensuring consistent product quality and reliability.
Market Competitiveness of Products:
Enterprises can enhance the competitiveness and market share of their products by evaluating performance and reliability through environmental simulation tests.
What safety issues should we pay attention to when doing constant temperature and humidity experiments?
High Temperature Caution:
During high-temperature experiments, the machine generates elevated temperatures within the chamber.
Safety Note: Exercise extreme caution when opening the chamber door during or after the experiment to prevent burns. Additionally, avoid contact with the high-temperature exhaust copper pipe from the freezer to prevent scalds.
Low Temperature Vigilance:
The machine achieves very low temperatures within the laboratory box during low-temperature experiments.
Safety Note: When opening the chamber door after a low-temperature experiment, take extra care to prevent frostbite. Be cautious of the extremely low temperatures generated, and handle materials with care to avoid any risk of frostbite.
Are you doing the daily maintenance of the constant temperature and humidity test chamber correctly?
- Constant temperature and humidity test box maintenance is very basic is to check whether the screws, parts, connecting lines, boards are loose, corrosion, missing, not solid, depression, cracks, etc., check gaskets, plugs, seals are aging, and deal with it in a timely manner.
- The compressor plays a very important role in the instrument, and expensive. Therefore, it is also important to do a good job of routine maintenance of the compressor. The work of the compressor requires a stable voltage, otherwise it is easy to cause damage. Therefore, a voltage stabilizer should be installed on the test box; the compressor cannot be switched on and off frequently. Each switch interval should be more than 15 minutes.
- If the equipment motor, turntable reducer does not run well or there is a murmur, debris should be removed and lubricated with oil.
- Regularly check whether there is leakage or stain deposition in the pump, pipeline and drainage box in the circulatory system, and carry out maintenance and cleaning.
- If the constant temperature and humidity test box is not used for a long time, the power supply should be cut off, clean and dry the box, placed in a dry, ventilated, no direct sunlight. Shall not cover the equipment with plastic to prevent dust from falling off, which may easily cause corrosion of parts. Even if not in use, it is best to energize once a month for about an hour each time to protect the compressor.
- In order to prevent parts from rusting, can be regularly wiped with oil; equipment should be added in a timely manner, shall not use the tap water with more impurities, should use pure water, distilled water and so on.
You can learn more about daily maintenance information from this blog: How To Maintain The ChiuVention CV340 AtmoExplorer Temperature And Humidity Chamber